Just like all machines daily centrifuge maintenance and cleaning is needed for the tool to operate properly. A couple of steps are needed to ensure the centrifuge runs in optimal performance with minimal centrifuge repairs with minimum human supervision as possible.
Daily cleaning of the inside surfaces of the centrifuge with a dilution of household bleach or an equivalent disinfectant is crucial. Portions of the centrifuge in contact with blood and other potentially infectious agent must be immediately cleaned when tube cracks or breakage ever happens.
All loose materials like plastic or broken glass must be carefully removed and disposed. Measured and rated speeds should not differ by more than five percent under specified conditions. Centrifuge speeds should periodically be checked using a reliable photoelectric tachometer relating with the College of American Pathologist (CAP) inspection guidelines Netapp Support Proper centrifuge maintenance and centrifuge repair guidelines requires that all centrifuge bowls should be cleaned by having an antibacterial solution. Rotor heads and buckets should be cleaned as well.
Using CAP guidelines , the accuracy of the centrifuge timer should also be checked and verified on a typical basis. Measured and programmed temperature should be within two degrees Celsius. The temperature of refrigerated centrifuges should be checked at least once a monthly in standardized conditions. If one does not keep to this task, then centrifuge repairs of machine parts may be required. Centrifuges with seven hundred to fourteen hundred running hours after minor repairs should be carried out, good maintenance and repair of centrifuges is an important guarantee for its safe operation. Machines running between thee thousand to four thousand work hours will demand an overhaul with repairs to any damaged drums, corroded shafts, rim fractures, and loose shells. Minor repairs must be performed to wash the bearing components and tighten all bolts. When doing centrifuge repairs , replace the screen, brake band, check the bearings, and replace the rubber shock absorbers.
Manufacturer’s instructions for lubrication, maintenance and replacement of brushes should be followed on a typical basis. Failure to displace worn brushes might cause the motor to fail and require replacement. All centrifuge maintenance function checks must be recorded and all corrective actions documented. A preventative maintenance schedule is the greatest means for customers to care for their equipment, without the headache of getting to remember to schedule service. It’s automatic, and you are able to select the month of the year that is most effective for you personally!